17 August 2018 (Friday) - D Dimer Calibration
I did a calibration on the D-Dimer assay today. Despite following the instructions things didn’t go according to plan. After a little struggling it turns out that although the device can store multiple reagent lot data, it only stores one calibration file. When the calibrator expiry date is reached the calibration and assay is automatically invalidated even though the calibration is still acceptable.
Therefore DO NOT have multiple calibrator information stored on-board.
I always say a day when you learn nothing is a day wasted…